Most Active Hardware Boards: Open the catalogue to page 5. Important product and driver updates Calibration reminders to keep your hardware in compliance Notifications about warranty expiration Simplified asset management for all your products on ni. Please take a moment to register your NI products to receive: Message 9 of I was wondering how I can fix this problem. Web-Based Configuration and Monitoring.
|Date Added:||22 November 2007|
|File Size:||51.55 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Or if your chassis is automatically detected on the network, it will appear under the Available Devices section.
Select a Web Site
Extended Warranties National Instruments designs and ccdaq-9178 all products to minimize failures, however unexpected failures can still occur.
Open the catalogue to page 4. It should be a driver problem, but I have already installed the latest version of driver i.
Message 3 of Please take a moment to register your NI products to receive:. Of the three computers you tried, is this the only computer where it is not recognized?
You must use these screws because they are the correct depth and thread for the panel. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
The NI desktop mounting kit includes two metal feet you can install on the sides of the CompactDAQ chassis for desktop use.
cDAQ – National Instruments
Use the following kits for each chassis:. The Online Industrial Exhibition. Open the catalogue to page 2.
cDAQ Support – National Instruments
Message 5 of Message 7 of Message 9 of Modules are available for a variety of sensor measurements including thermocouples, RTDs, strain gages, load and pressure transducers, torque cells, accelerometers, flow meters, and microphones. You can panel mount cDAQ with either panhead machine screws or M3.
Connect Sensors or Signals Next. Because your is working when connecting to other computers, I think that the issue is related to your computer.
Two keyholes are located on cDAQ for mounting it to a panel or wall. Thanks, I’m gonna do what you told. Alternatively, you can use the NI Network Browser located in the NI start menu folder on your host computer to find chassis on ddaq-9178 local subnet. To mount the chassis on a panel, align the chassis on the panel mount accessory.
Thanks, I’m going to test it on aonther PC. Can you compare this with your other computer to verify that the device is not being recognized by Windows? Message 6 of I connected the power cable to the NI Open the catalogue to page 5.
You can access these counters through an installed, hardware-timed digital module such as the NI or NI for applications that involve quadrature encoders, PWM, event counting, pulse train generation, and period or frequency measurement. Select this device and click Add Selected Devices. Ring Lug Attached to Chassis Ground. The cDAQ chassis is shipped with the following: